The surgeon control panel should meet Electrical safety codes for high and low voltage systems, wired to the current IEE regulations. The room Surgeon control panel should be modifiable with changing technology and equipment in operating environments. Control Panel will be user-friendly and easy of operating and maintaining purpose.The panel should be Membrane type; configured to incorporate all the services that operation room staff required.The should be made with a superior quality UV resistance membrane with sterilization feature. The Panel should contain 6 or 9 service tiles as below: 1.Time Day Clock. 2.Time Elapse Day Clock. 3. General Lighting System. 4.Medical Gas Alarm Panel. 5. Hand Free Telephone set with memory. 6.Temperature and humidity indicator with the controller. 7.HEPA Filter Status module. 8. Room Pressure Indicator. 9. Music control. Time day clock should be digital type and clocks have high brightness characters. Time Elapsed Day Clock should be digital type and clocks having high brightness characters.Temperature indicator should indicate the room temperature which should be connected to the local pressure switches of Air Conditioning System. Indicators should be digital type and clocks having high brightness characters, not less than 30 mm in height. The Central Lighting system should incorporate all the necessary controls of the lighting. the system inside the theatre. The medical gas alarm should indicate High, Normal and Low gas Pressure for each gas service present in the operating system and should have an audible buzzer with a mute facility. Pressure sensors should be connected to MGPS for monitoring the pressures. A hand-free set Telephone System should be incorporated into the panel with the memory-type card.The control panel should be designed for front access only. All internal wires should be marked with plastic Ferrule-type cable markers for ease of identification.
Developed and Managed by Infocom Network Private Limited.